YTS61 oil property sensor uses the world’s advanced piezoelectric resonant MEMS components, through the integrated high-precision signal sampling and processing unit, combined with advanced algorithms, can automatically detect the moisture, density, viscosity, dielectric constant, water activity and temperature. At the same time, 20 ° C density, kinematic viscosity, viscosity index；40 ° C viscosity, 100 ° C viscosity and other indicators can also be achieved through a powerful internal calculation program.
- Imported probes, accurate measurement of viscosity, density and temperature up to eight parameters output online monitoring sensor;
- Fast response, the data refresh frequency every second;
- The outer casing is made of stainless steel and has excellent chemical and pressure resistance. The measurement is not affected by external vibration and is suitable for on-line monitoring of robustness requirements;
- No moving parts, no consumable parts, effective service life up to more than 10 years;
- Compact structure, easy system integration, on-site calibration of integrated modules;
- CE, ASTM1657 and other certification and testing standards.
YTS61 is widely applied to online monitor the viscosity and density of lubricating oil, hydraulic oil, gear oil, gasoline, diesel oil, jet fuel, kerosene, biodiesel, ethanol, chemical reagent and coating ink.
YS61 is applied to food processing, laboratory, fuel oil storage and transportation ; Oil refining, wind power and other power industries; Aviation, navigation, railway and other transportation industry;Drilling platform and other maritime engineer, mechanical engineering equipment,transportation vehicle,fuel storage and transportation vehicle, oil handling equipment, chemical test and analysis,oil handling and filter system; Coating, painting and other printing industry, medical equipment,etc.
Installation steps (pictured above):
Step1:Insert the connecting wire into the terminal of the sensor, turn the threaded part of the connecting wire and tighten it;
Step2:Connect the sensor to the pipe, and the connection method is thread rotation.·
Note:Ensure that the oil path is from bottom to top during installation.